Medical device alert - Philips HeartStart XL+ Defibrillator
7 November 2019
The MHRA is alerting dentists following a notification from Philips that the HeartStart XL+ Defibrillator/Monitor (Model number 861290) may fail to turn on or unexpectedly attempt to restart, and this failed restart may result in a delay if the defibrillator/monitor is needed for immediate use.
The device is safe to use and can remain in service if it does not exhibit any of the behaviours described above. If it does reboot itself, or fails to start, remove the device from service and contact Philips.
Philips will perform a system software upgrade and replace the Processor PCA that
contains the SOM module that is subject to failure.
Until the required parts are available and this service is performed, the device should remain removed from service, regardless of whether it is subsequently restarted successfully.
For information or support, contact your local Philips representative or the UK Philips Customer Care Service Centre on 0870 532 9741.
BDA advisory services
Our advisory services team offers advice to BDA members on a huge range of issues, such as employment law, health and safety, the NHS, business support and regulatory inspections. We also provide employment representation, an associate contract-checking service, mediation services, tribunal support and consultancy services.
With each new member, our voice and our influence grows.
Add your voice, join today.